CIBIO® Ovulation Test Cassette
One Step LH Test Cassette is an in vitro diagnostic test device intended for the rapid determination of human Luteinzing Hormones (LH ) in urine to predict when there is a LH surge, and in turn, when you are likely to ovulate. You can easily carry out yourself.
It is intended for over-the-counter (OTC) use. For in vitro diagnostic use only.
|Product Name||LH Ovulation Test Cassette|
|Brand Name||CIBIO® or OEM|
|Kits content||1. Test cassette|
|4. Instruction for use|
|Cut-off Level||20 or 25 mIU/ml|
|Cassette Size||4.0 mm|
|Certificates||CE 0123, ISO 13485, FDA, MHRA, CFS/ FSC|
|Storage Temperature||4-30 degree centigrade|
|Shelf Life||24-36 Months|
|OEM Services||Logo, pouch, box, insert|
|Loading Port||Guangzhou, Shenzhen|
How to Do the Test?
- Remove a Testing Device from the foil pouch by tearing at the notch and place it on a level surface.
- Holding a sample dropper vertically, add 2-3 drops of the urine specimen to the sample well (with an arrow marked).
- Read the result in five minutes. Positive results may sometimes appear as early as one minute but you should wait the full 5 minutes before determining a negative result. Do not read results after more than 5 minutes.
How to read the results?
Negative (No LH Surge) :
Only one color band appears on the control region, or the test band appears but is lighter than the control band. This means there is no LH surge.
Positive (LH Surge) :
If two color bands are visible, and the test band is equal to or darker than the control band, one will probably ovulate in the next 24-48 hours. If trying to get pregnant, the best time to have intercourse is after 24 but before 48 hours.
No visible band at all, or there is a visible band only in the test region and not in the control region. Repeat with a new test kit.